We sought to determine the safety and feasibility of percutaneous MRI-guided cryotherapy in the care of patients with refractory or painful metastatic lesions of soft tissue and bone adjacent to critical structures.
MATERIALS AND METHODS:
Twenty-seven biopsy-proven metastatic lesions of soft tissue (n = 17) and bone (n = 10) in 22 patients (15 men, seven women; age range, 24-85 years) were managed with MRI-guided percutaneous cryotherapy. The mean lesion diameter was 5.2 cm. Each lesion was adjacent to or encasing one or more critical structures, including bowel, bladder, and major blood vessels. A 0.5-T open interventional MRI system was used for cryoprobe placement and ice-ball monitoring. Complications were assessed for all treatments. CT or MRI was used to determine local control of 21 tumors. Pain palliation was assessed clinically in 19 cases. The mean follow-up period was 19.5 weeks.
Twenty-two (81%) of 27 tumors were managed without injury to adjacent critical structures. Two patients had transient lower extremity numbness, and two had both urinary retention and transient lower extremity paresthesia. One patient had chronic serous vaginal discharge, and one sustained a femoral neck fracture at the ablation site 6 weeks after treatment. Thirteen (62%) of the 21 tumors for which follow-up information was available either remained the same size as before treatment or regressed. Eight tumors progressed (mean local progression-free interval, 5.6 months; range, 3-18 months). Pain was palliated in 17 of 19 patients; six of the 17 experienced complete relief, and 11 had partial relief.
MRI-guided percutaneous cryotherapy for metastatic lesions of soft tissue and bone adjacent to critical structures is safe and can provide local tumor control and pain relief in most patients.
This contribution presents finite element computation of the deformation field within the brain during craniotomy-induced brain shift. The results were used to illustrate the capabilities of non-linear (i.e. accounting for both geometric and material non-linearities) finite element analysis in non-rigid registration of pre- and intra-operative magnetic resonance images of the brain. We used patient-specific hexahedron-dominant finite element mesh, together with realistic material properties for the brain tissue and appropriate contact conditions at boundaries. The model was loaded by the enforced motion of nodes (i.e. through prescribed motion of a boundary) at the brain surface in the craniotomy area. We suggest using explicit time-integration scheme for discretised equations of motion, as the computational times are much shorter and accuracy, for practical purposes, the same as in the case of implicit integration schemes. Application of the computed deformation field to register (i.e. align) the pre-operative images with the intra-operative ones indicated that the model very accurately predicts the displacements of the tumour and the lateral ventricles even for limited information about the brain surface deformation. The prediction accuracy improves when information about deformation of not only exposed (during craniotomy) but also unexposed parts of the brain surface is used when prescribing loading. However, it appears that the accuracy achieved using information only about the deformation of the exposed surface, that can be determined without intra-operative imaging, is acceptable. The presented results show that non-linear biomechanical models can complement medical image processing techniques when conducting non-rigid registration. Important advantage of such models over the previously used linear ones is that they do not require unrealistic assumptions that brain deformations are infinitesimally small and brain stress-strain relationship is linear.
Evidence indicates that sleep after learning is critical for the subsequent consolidation of human memory. Whether sleep before learning is equally essential for the initial formation of new memories, however, remains an open question. We report that a single night of sleep deprivation produces a significant deficit in hippocampal activity during episodic memory encoding, resulting in worse subsequent retention. Furthermore, these hippocampal impairments instantiate a different pattern of functional connectivity in basic alertness networks of the brainstem and thalamus. We also find that unique prefrontal regions predict the success of encoding for sleep-deprived individuals relative to those who have slept normally. These results demonstrate that an absence of prior sleep substantially compromises the neural and behavioral capacity for committing new experiences to memory. It therefore appears that sleep before learning is critical in preparing the human brain for next-day memory formation-a worrying finding considering society's increasing erosion of sleep time.
Malformations of the cerebral cortex are recognized as a common cause of developmental delay, neurological deficits, mental retardation and epilepsy. Currently, the diagnosis of cerebral cortical malformations is based on a subjective interpretation of neuroimaging characteristics of the cerebral gray matter and underlying white matter. There is no automated system for aiding the observer in making the diagnosis of a cortical malformation. In this paper a fuzzy rule-based system is proposed as a solution for this problem. The system collects the available expert knowledge about cortical malformations and assists the medical observer in arriving at a correct diagnosis. Moreover, the system allows the study of the influence of the various factors that take part in the decision. The evaluation of the system has been carried out by comparing the automated diagnostic algorithm with known case examples of various malformations due to abnormal cortical organization. An exhaustive evaluation of the system by comparison with published cases and a ROC analysis is presented in the paper.
OBJECTIVES: The determination of eloquent cortex is essential when planning neurosurgical approaches to brain lesions. This study examined the abilities of medical personnel of various backgrounds to predict the location of functional cortex using anatomical information provided by MR imaging.
PATIENTS AND METHODS: Neurosurgeons, neuroscientists, neuroradiologists, medical students and MR technologists viewed anatomical MR images acquired from patients with brain tumors and healthy controls. These five groups of raters were then asked to locate the primary motor hand, supplementary motor and primary auditory areas and their predictions were compared to fMRI data acquired from the same subjects.
RESULTS: The overall mean distance from the center of the fMRI activation was 2.38 cm. The neuroscientists performed the best and MR technologists performed the worst (mean distance from center of 1.83 and 3.04 cm, respectively, p<0.05). The difference between patients and controls was not significant. The mean distance by ROI was primary motor hand 2.03 cm, auditory area 2.06 cm and supplementary motor area 3.18 cm (p<0.05). Raters also performed best in the medial-lateral direction, compared to superior-inferior and anterior-posterior directions (mean distances from center 0.42, 1.04 and 1.81 cm, respectively). Finally, the approximate minimum fields of view necessary to capture the entire fMRI activations using the raters' predictions ranged from 5 to 15 cm, or 3 to 12 cm larger than the fMRI activations.
CONCLUSION: Medical personnel of various training perform poorly when using only anatomical information to predict the location of functional areas of cortex.
A method is described for detecting scattering in two-dimensions using an unfocused ultrasound field created from a continuously driven source array. The frequency of each element on the array is unique, resulting in a field that is highly variant as a function of both time and position. The scattered signal is then received by a single receiving line. The method, as currently written, is valid under the first order Born approximation. To demonstrate the approach, a series of simulations within the frequency range of 0.10-1.25 MHz are performed and compared with a simulated B-Scan in the same frequency range. The method is found to be superior in resolving closely spaced objects, discerning 1.4 mm separation in the radial and 0.5-mm separation in the axial direction. The method was also better able to determine object size, resolving scatters less than 10% of wavelength associated with the center frequency.
The investigation of the reproducibility in functional MRI (fMRI) is an important step in the quantification and analysis of paradigm-related brain activation. This article reports on reproducibility of cortical activation characterized by repeated fMRI runs (10 times) during the performance of a motor imagery and a passive auditory stimulation as a control task. Two parameters, the size of activation and BOLD signal contrast, were measured from regions-of-interest for 10 subjects across different threshold conditions. The variability of these parameters was normalized with respect to the mean obtained from 10 runs, and represented as the intrasession variability. It was found that the variability was significantly lower in the measurement of BOLD signal contrast as compared to the measurement of the size of activation. The variability of the activation volume measurement was greater in the motor imagery task than in the auditory tasks across all thresholds. This task-dependent difference was not apparent from the measurement of the BOLD signal contrast. The presence of threshold dependence in the variability measurement was also examined, but no such dependency was found. The results suggest that a measurement of BOLD signal itself is a more reliable indicator of paradigm-related brain activation during repeated fMRI scans.
In prostate cancer treatment, there is a move toward targeted interventions for biopsy and therapy, which has precipitated the need for precise image-guided methods for needle placement. This paper describes an integrated system for planning and performing percutaneous procedures with robotic assistance under MRI guidance. A graphical planning interface allows the physician to specify the set of desired needle trajectories, based on anatomical structures and lesions observed in the patient's registered pre-operative and pre-procedural MR images, immediately prior to the intervention in an open-bore MRI scanner. All image-space coordinates are automatically computed, and are used to position a needle guide by means of an MRI-compatible robotic manipulator, thus avoiding the limitations of the traditional fixed needle template. Automatic alignment of real-time intra-operative images aids visualization of the needle as it is manually inserted through the guide. Results from in-scanner phantom experiments are provided.
We have developed a method to use low-intensity focused ultrasound pulses combined with an ultrasound contrast agent to produce temporary blood-brain barrier disruption (BBBD). This method could provide a means for the targeted delivery of drugs or imaging agents into the brain. In all our previous work, we used Optison as the ultrasound contrast agent. The purpose of this study was to test the feasibility of using the contrast agent Definity for BBBD. A total of 36 non-overlapping locations were sonicated through a craniotomy in experiments in the brains of nine rabbits (four locations per rabbit; ultrasound [US] frequency: 0.69 MHz; burst: 10 ms; pulse repetition frequency (PRF): 1 Hz; duration: 20 s). The peak negative pressure amplitude ranged from 0.2 to 1.5 MPa. An additional 11 locations were sonicated using Optison at pressure amplitude of 0.5 MPa. Definity and Optison dosages were the same as those used clinically for ultrasound imaging: 10 and 50 microl/kg, respectively. The probability for BBBD (determined using MRI contrast agent enhancement) as a function of pressure amplitude was similar to that found earlier with Optison. For both agents, the probability was estimated to be 50% at 0.4 MPa using probit regression. Histologic examination revealed small, isolated areas of extravasated erythrocytes in some locations. At 0.8 MPa and higher, these areas were sometimes accompanied by tiny (dimensions of 100 microm or less) regions of damaged brain parenchyma. The magnitude of the BBBD was larger with Optison than with Definity at 0.5 MPa (signal enhancement: 13.3% +/- 4.4% vs. 8.4% +/- 4.9%; p = 0.04). In addition, more areas with extravasated erythrocytes were observed with Optison (5.0 +/- 3.5 vs. 1.4 +/- 1.9 areas with extravasation in histology section with largest effect; p = 0.03). We concluded that BBBD is possible using Definity at the dosage of contrast agent and the acoustic parameters tested in this study. The probability for BBBD as a function of pressure amplitude and the type of acute tissue effects were similar to what has been observed using Optison. However, under the experimental conditions used in this study, Optison produced a larger effect for the same acoustic pressure amplitude.
RATIONALE AND OBJECTIVES: To perform a retrospective, quantitative assessment of the anatomic relationship between intra-axial, supratentorial, primary brain tumors, and adjacent white matter fiber tracts based on anatomic and diffusion tensor magnetic resonance imaging (MRI). We hypothesized that white matter infiltration may be common among different types of tumor.
MATERIAL AND METHODS: Preoperative, anatomic (T1- and T2-weighted), and LINESCAN diffusion tensor MRI were obtained in 12 patients harboring supratentorial gliomas (World Health Organization [WHO] Grades II and III). The two imaging modalities were rigidly registered. The tumors were manually segmented from the T1- and T2-weighted MRI, and their volume calculated. A three-dimensional tractography was performed in each case. A second segmentation and volume measurement was performed on the tumor regions intersecting adjacent white matter fiber tracts. Statistical methods included summary statistics to examine the fraction of tumor volume infiltrating adjacent white matter.
RESULTS: There were five patients with low-grade oligodendroglioma (WHO Grade II), one with low-grade mixed oligoastrocytoma (WHO Grade II), one with ganglioglioma, two with low-grade astrocytoma (WHO Grade II), and three with anaplastic astrocytoma (WHO Grade III). We identified white matter tracts infiltrated by tumor in all 12 cases. The median tumor volume (+/- standard deviation) in our patient population was 42.5 +/- 28.9 mL. The median tumor volume (+/- standard deviation) infiltrating white matter fiber tracts was 5.2 +/- 9.9 mL. The median percentage of tumor volume infiltrating white matter fiber tracts was 21.4% +/- 9.7%.
CONCLUSIONS: The information provided by diffusion tensor imaging combined with anatomic MRI might be useful for neurosurgical planning and intraoperative guidance. Our results confirm previous reports that extensive white matter infiltration by primary brain tumors is a common occurrence. However, prospective, large population studies are required to definitively clarify this issue, and how infiltration relates to histologic tumor type, tumor size, and location.
Tissue water molecules reside in different biophysical compartments. For example, water molecules in the vasculature reside for variable periods of time within arteries, arterioles, capillaries, venuoles and veins, and may be within blood cells or blood plasma. Water molecules outside of the vasculature, in the extravascular space, reside, for a time, either within cells or within the interstitial space between cells. Within these different compartments, different types of microscopic motion that water molecules may experience have been identified and discussed. These range from Brownian diffusion to more coherent flow over the time scales relevant to functional magnetic resonance imaging (fMRI) experiments, on the order of several 10s of milliseconds. How these different types of motion are reflected in magnetic resonance imaging (MRI) methods developed for "diffusion" imaging studies has been an ongoing and active area of research. Here we briefly review the ideas that have developed regarding these motions within the context of modern "diffusion" imaging techniques and, in particular, how they have been accessed in attempts to further our understanding of the various contributions to the fMRI signal changes sought in studies of human brain activation.
The integration of medical devices with software applications is crucial for image-guided medical applications. This work describes a general device interface that has been designed for high-frequency streaming of multi-modal events, thus providing maximum performance and flexibility for such applications. Several sample applications and performance tests are provided to demonstrate the usability of the concept.
Regional investigations of newborn MRI are important to understand the appearance and consequences of early brain injury. Previously, regionalization in neonates has been achieved with a Talairach parcellation, using internal landmarks of the brain. Non-synostotic dolichocephaly defines a bi-temporal narrowing of the preterm infant's head caused by pressure on the immature skull. The impact of dolichocephaly on brain shape and regional brain shift, which may compromise the validity of the parcellation scheme, has not yet been investigated. Twenty-four preterm and 20 fullterm infants were scanned at term equivalent. Skull shapes were investigated by cephalometric measurements and population registration. Brain tissue volumes were calculated to rule out brain injury underlying skull shape differences. The position of Talairach landmarks was evaluated. Cortical structures were segmented to determine a positional shift between both groups. The preterm group displayed dolichocephalic head shapes and had similar brain volumes compared to the mesocephalic fullterm group. In preterm infants, Talairach landmarks were consistently positioned relative to each other and to the skull base, but were displaced with regard to the calvarium. The frontal and superior region was enlarged; central and temporal gyri and sulci were shifted comparing preterm and fullterm infants. We found that, in healthy preterm infants, dolichocephaly led to a shift of cortical structures, but did not influence deep brain structures. We concluded that the validity of a Talairach parcellation scheme is compromised and may lead to a miscalculation of regional brain volumes and inconsistent parcel contents when comparing infant populations with divergent head shapes.
This paper presents a novel multiscale shape representation and segmentation algorithm based on the spherical wavelet transform. This work is motivated by the need to compactly and accurately encode variations at multiple scales in the shape representation in order to drive the segmentation and shape analysis of deep brain structures, such as the caudate nucleus or the hippocampus. Our proposed shape representation can be optimized to compactly encode shape variations in a population at the needed scale and spatial locations, enabling the construction of more descriptive, nonglobal, nonuniform shape probability priors to be included in the segmentation and shape analysis framework. In particular, this representation addresses the shortcomings of techniques that learn a global shape prior at a single scale of analysis and cannot represent fine, local variations in a population of shapes in the presence of a limited dataset. Specifically, our technique defines a multiscale parametric model of surfaces belonging to the same population using a compact set of spherical wavelets targeted to that population. We further refine the shape representation by separating into groups wavelet coefficients that describe independent global and/or local biological variations in the population, using spectral graph partitioning. We then learn a prior probability distribution induced over each group to explicitly encode these variations at different scales and spatial locations. Based on this representation, we derive a parametric active surface evolution using the multiscale prior coefficients as parameters for our optimization procedure to naturally include the prior for segmentation. Additionally, the optimization method can be applied in a coarse-to-fine manner. We apply our algorithm to two different brain structures, the caudate nucleus and the hippocampus, of interest in the study of schizophrenia. We show: 1) a reconstruction task of a test set to validate the expressiveness of our multiscale prior and 2) a segmentation task. In the reconstruction task, our results show that for a given training set size, our algorithm significantly improves the approximation of shapes in a testing set over the Point Distribution Model, which tends to oversmooth data. In the segmentation task, our validation shows our algorithm is computationally efficient and outperforms the Active Shape Model algorithm, by capturing finer shape details.
A system was developed for real-time electrocardiogram (ECG) analysis and artifact correction during magnetic resonance (MR) scanning, to improve patient monitoring and triggering of MR data acquisitions. Based on the assumption that artifact production by magnetic field gradient switching represents a linear time invariant process, a noise cancellation (NC) method is applied to ECG artifact linear prediction. This linear prediction is performed using a digital finite impulse response (FIR) matrix, that is computed employing ECG and gradient waveforms recorded during a training scan. The FIR filters are used during further scanning to predict artifacts by convolution of the gradient waveforms. Subtracting the artifacts from the raw ECG signal produces the correction with minimal delay. Validation of the system was performed both off-line, using prerecorded signals, and under actual examination conditions. The method is implemented using a specially designed Signal Analyzer and Event Controller (SAEC) computer and electronics. Real-time operation was demonstrated at 1 kHz with a delay of only 1 ms introduced by the processing. The system opens the possibility of automatic monitoring algorithms for electrophysiological signals in the MR environment.
OBJECTIVE: Virtual endoscopic simulations using volume rendering (VR) have been proposed as a tool for training and understanding intraventricular anatomy. It is not known whether surface rendering (SR), an alternative to VR, can visualize intraventricular and subependymal structures better and thus making the virtual endoscope more useful for simulating the intraventricular endoscopy. We sought to develop SR-virtual endoscopy and compared the visibility of anatomical structures in SR and VR using retrospective cases.
MATERIALS AND METHODS: Fourteen patients who underwent endoscopic intraventricular surgery of third ventricle enrolled the study. SR-virtual endoscopy module was developed in open-source software 3D Slicer and virtual endoscopic scenes from the retrospective cases were created. VR virtual endoscopy of the same cases was prepared in commercial software. Three neurosurgeons scored the visibility of substructures in lateral and third ventricle, arteries, cranial nerves, and other lesions Results: We found that VR and SR-virtual endoscopy performed similarly in visualization of substructures in lateral and third ventricle (not significant statistically). However, the SR was statistically significantly better in visualizing subependymal arteries, cranial nerves, and other lesions (p<0.05, respectively).
CONCLUSIONS: We concluded that SR-virtual endoscopy is a promising tool to visualize critical anatomical structures in simulated endoscopic intraventricular surgery. The results lead us to propose a hybrid technique of volume and surface rendering to balance the strength of surface rendering alone in visualizing arteries, nerves and lesions, with fast volume rendering of third and lateral ventricles.
The clinical application of chemotherapy to brain tumors has been severely limited because antitumor agents are typically unable to penetrate an intact blood-brain barrier (BBB). Although doxorubicin (DOX) has been named as a strong candidate for chemotherapy of the central nervous system (CNS), the BBB often prevents cytotoxic levels from being achieved. In this study, we demonstrate a noninvasive method for the targeted delivery of DOX through the BBB, such that drug levels shown to be therapeutic in human tumors are achieved in the normal rat brain. Using MRI-guided focused ultrasound with preformed microbubbles (Optison) to locally disrupt the BBB and systemic administration of DOX, we achieved DOX concentrations of 886 +/- 327 ng/g tissue in the brain with minimal tissue effects. Tissue DOX concentrations of up to 5,366 +/- 659 ng/g tissue were achieved with higher Optison doses, but with more significant tissue damage. In contrast, DOX accumulation in nontargeted contralateral brain tissue remained significantly lower for all paired samples (p < 0.001). These results suggest that targeted delivery by focused ultrasound may render DOX chemotherapy a viable treatment option against CNS tumors, despite previous accessibility limitations. In addition, MRI signal enhancement in the sonicated region correlated strongly with tissue DOX concentration (r = 0.87), suggesting that contrast-enhanced MRI could perhaps indicate drug penetration during image-guided interventions. Our technique using MRI-guided focused ultrasound to achieve therapeutic levels of DOX in the brain offers a large step forward in the use of chemotherapy to treat patients with CNS malignancies.
PURPOSE: To prospectively assess patient response (after 12 months) to magnetic resonance (MR) imaging-guided focused ultrasound surgery in treatment of uterine leiomyomas by using two treatment protocols.
MATERIALS AND METHODS: This prospective clinical trial was approved by institutional review boards and was HIPAA compliant. After giving informed consent, patients with symptomatic leiomyomas were consecutively enrolled and treated at one of five U.S. centers by using an original or a modified protocol. Outcomes were assessed with the symptom severity score (SSS) obtained at baseline and 3, 6, and 12 months after treatment. Adverse events (AEs) were recorded. Statistical analysis included Student t test, Fisher exact test, analysis of covariance, Spearman correlation, and logistic regression.
RESULTS: One hundred sixty patients had a mean SSS of 62.1 +/- 16.3 (standard deviation) at baseline, which decreased to 35.5 +/- 19.5 at 3 months (P<.001) and to 32.3 +/- 19.8 at 6 months (P<.001) and was 32.7 +/- 21.0 at 12 months (P<.001). Ninety-six patients (mean age, 46.0 years +/- 4.6) were treated with an original protocol, and 64 (mean age, 45.9 years +/- 3.9) were treated with a modified protocol. Patients in the modified group had a significantly greater SSS decrease at 3 months (P=.037) than those in the original group, and 73% of those in the original group and 91% of those in the modified group reported a significant decrease in SSS (of 10 points or greater) at 12 months. No serious AEs were recorded. Fewer AEs were reported in the modified group than in the original group (25% vs 13% reporting no event). Of evaluable patients, fewer in the modified group chose alternative treatment (28%) than in the original group (37%).
CONCLUSION: MR imaging-guided focused ultrasound surgery results in symptomatic improvement, sustained to 12 months after treatment. Treatment with a modified protocol results in greater clinical effectiveness and fewer AEs.
We describe a new approach for estimating the posterior probability of tissue labels. Conventional likelihood models are combined with a curve length prior on boundaries, and an approximate posterior distribution on labels is sought via the Mean Field approach. Optimizing the resulting estimator by gradient descent leads to a level set style algorithm where the level set functions are the logarithm-of-odds encoding of the posterior label probabilities in an unconstrained linear vector space. Applications with more than two labels are easily accommodated. The label assignment is accomplished by the Maximum A Posteriori rule, so there are no problems of "overlap" or "vacuum". We test the method on synthetic images with additive noise. In addition, we segment a magnetic resonance scan into the major brain compartments and subcortical structures.
Patient-mounted needle guide devices for percutaneous ablation are vulnerable to patient motion. The objective of this study is to develop and evaluate a software system for an MRI-compatible patient-mounted needle guide device that can adaptively compensate for displacement of the device due to patient motion using a novel image-based automatic device-to-image registration technique. We have developed a software system for an MRI-compatible patient-mounted needle guide device for percutaneous ablation. It features fully-automated image-based device-to-image registration to track the device position, and a device controller to adjust the needle trajectory to compensate for the displacement of the device. We performed: (a) a phantom study using a clinical MR scanner to evaluate registration performance; (b) simulations using intraoperative time-series MR data acquired in 20 clinical cases of MRI-guided renal cryoablations to assess its impact on motion compensation; and (c) a pilot clinical study in three patients to test its feasibility during the clinical procedure. FRE, TRE, and success rate of device-to-image registration were [Formula: see text] mm, [Formula: see text] mm, and 98.3% for the phantom images. The simulation study showed that the motion compensation reduced the targeting error for needle placement from 8.2 mm to 5.4 mm (p < 0.0005) in patients under general anesthesia (GA), and from 14.4 mm to 10.0 mm ([Formula: see text]) in patients under monitored anesthesia care (MAC). The pilot study showed that the software registered the device successfully in a clinical setting. Our simulation study demonstrated that the software system could significantly improve targeting accuracy in patients treated under both MAC and GA. Intraprocedural image-based device-to-image registration was feasible.
PURPOSE: To develop and evaluate an approach to estimate the respiratory-induced motion of lesions in the chest and abdomen. MATERIALS AND METHODS: The proposed approach uses the motion of an initial reference needle inserted into a moving organ to estimate the lesion (target) displacement that is caused by respiration. The needles position is measured using an inertial measurement unit (IMU) sensor externally attached to the hub of an initially placed reference needle. Data obtained from the IMU sensor and the target motion are used to train a learning-based approach to estimate the position of the moving target. An experimental platform was designed to mimic respiratory motion of the liver. Liver motion profiles of human subjects provided inputs to the experimental platform. Variables including the insertion angle, target depth, target motion velocity and target proximity to the reference needle were evaluated by measuring the error of the estimated target position and processing time. RESULTS: The mean error of estimation of the target position ranged between 0.86 and 1.29 mm. The processing maximum training and testing time was 5 ms which is suitable for real-time target motion estimation using the needle position sensor. CONCLUSION: The external motion of an initially placed reference needle inserted into a moving organ can be used as a surrogate, measurable and accessible signal to estimate in real-time the position of a moving target caused by respiration; this technique could then be used to guide the placement of subsequently inserted needles directly into the target.
Brain shift during tumor resection compromises the spatial validity of registered preoperative imaging data that is critical to image-guided procedures. One current clinical solution to mitigate the effects is to reimage using intraoperative magnetic resonance (iMR) imaging. Although iMR has demonstrated benefits in accounting for preoperative-to-intraoperative tissue changes, its cost and encumbrance have limited its widespread adoption. While iMR will likely continue to be employed for challenging cases, a cost-effective model-based brain shift compensation strategy is desirable as a complementary technology for standard resections. We performed a retrospective study of [Formula: see text] tumor resection cases, comparing iMR measurements with intraoperative brain shift compensation predicted by our model-based strategy, driven by sparse intraoperative cortical surface data. For quantitative assessment, homologous subsurface targets near the tumors were selected on preoperative MR and iMR images. Once rigidly registered, intraoperative shift measurements were determined and subsequently compared to model-predicted counterparts as estimated by the brain shift correction framework. When considering moderate and high shift ([Formula: see text], [Formula: see text] measurements per case), the alignment error due to brain shift reduced from [Formula: see text] to [Formula: see text], representing [Formula: see text] correction. These first steps toward validation are promising for model-based strategies.
OBJECTIVE: The purpose of this article is to report our intermediate to long-term outcomes with image-guided percutaneous hepatic tumor cryoablation and to evaluate its technical success, technique efficacy, local tumor progression, and adverse event rate. MATERIALS AND METHODS: Between 1998 and 2014, 299 hepatic tumors (243 metastases and 56 primary tumors; mean diameter, 2.5 cm; median diameter, 2.2 cm; range, 0.3-7.8 cm) in 186 patients (95 women; mean age, 60.9 years; range, 29-88 years) underwent cryoablation during 236 procedures using CT (n = 126), MRI (n = 100), or PET/CT (n = 10) guidance. Technical success, technique efficacy at 3 months, local tumor progression (mean follow-up, 2.5 years; range, 2 months to 14.6 years), and adverse event rates were calculated. RESULTS: The technical success rate was 94.6% (279/295). The technique efficacy rate was 89.5% (231/258) and was greater for tumors smaller than 4 cm (93.4%; 213/228) than for larger tumors (60.0%; 18/30) (p < 0.0001). Local tumor progression occurred in 23.3% (60/258) of tumors and was significantly more common after the treatment of tumors 4 cm or larger (63.3%; 19/30) compared with smaller tumors (18.0%; 41/228) (p < 0.0001). Adverse events followed 33.8% (80/236) of procedures and were grade 3-5 in 10.6% (25/236) of cases. Grade 3 or greater adverse events more commonly followed the treatment of larger tumors (19.5%; 8/41) compared with smaller tumors (8.7%; 17/195) (p = 0.04). CONCLUSION: Image-guided percutaneous cryoablation of hepatic tumors is efficacious; however, tumors smaller than 4 cm are more likely to be treated successfully and without an adverse event.
OBJECTIVE: We report nine consecutive percutaneous image-guided cryoablation procedures of head and neck tumors in seven patients (four men and three women; mean age, 68 years; age range, 50-78 years). Ablation of the entire tumor for local control or ablation of a region of tumor for pain relief or preservation of function was achieved in eight of nine procedures. One patient experienced intraprocedural bradycardia, and another developed a neopharyngeal abscess. There were no deaths, permanent neurologic or functional deficits, vascular complications, or adverse cosmetic sequelae due to the procedures. CONCLUSION: Percutaneous image-guided cryoablation offers a potentially less morbid minimally invasive treatment option than salvage head and neck surgery. The complications that we encountered may be avoidable with increased experience. Further work is needed to continue improving the safety and efficacy of cryoablation of head and neck tumors and to continue expanding the use of cryoablation in patients with head and neck tumors that cannot be treated surgically.