Pituitary adenomas are relatively common intracranial neoplasms that are frequently treated with surgical resection. Rapid visualization of pituitary tissue remains a challenge as current techniques either produce little to no information on hormone-secreting function or are too slow to practically aid in intraoperative or even perioperative decision-making. Matrix-assisted laser desorption/ionization mass spectrometry imaging (MALDI MSI) represents a powerful method by which molecular maps of tissue samples can be created, yielding a two-dimensional representation of the expression patterns of small molecules and proteins from biologic samples. In this chapter, we review the use of MALDI MSI, its application to the characterization of the pituitary gland, and its potential applications for guiding the management of pituitary adenomas.
OBJECTIVE: The purpose was to compare local control (LC), overall survival (OS) and dose to the organs at risk (OAR) in women with locally advanced cervical cancer treated with MR-guided versus CT-guided interstitial brachytherapy (BT). METHODS: 56 patients (29 MR, 27 CT) were treated with high-dose-rate (HDR) interstitial BT between 2005-2015. The MR patients had been prospectively enrolled on a Phase II clinical trial. Data were analyzed using Kaplan-Meier (K-M) and Cox proportional hazards statistical modeling in JMP® & R®. RESULTS: Median follow-up time was 19.7months (MR group) and 18.4months (CT group). There were no statistically significant differences in patient age at diagnosis, histology, percent with tumor size >4cm, grade, FIGO stage or lymph node involvement between the groups. Patients in the MR group had more lymphovascular involvement compared to patients in the CT group (p<0.01). When evaluating plans generated, there were no statistically significant differences in median cumulative dose to the high-risk clinical target volume or the OAR. 2-year K-M LC rates for MR-based and CT-based treatments were 96% and 87%, respectively (log-rank p=0.65). At 2years, OS was significantly better in the MR-guided cohort (84% vs. 56%, p=0.036). On multivariate analysis, squamous histology was associated with longer OS (HR 0.23, 95% CI 0.07-0.72) in a model with MR BT (HR 0.35, 95% CI 0.08-1.18). There was no difference in toxicities between CT and MR BT. CONCLUSION: In this population of locally advanced cervical-cancer patients, MR-guided HDR BT resulted in estimated 96% 2-year local control and excellent survival and toxicity rates.
PURPOSE: To determine the accuracy of in-bore transperineal 3-Tesla (T) magnetic resonance (MR) imaging-guided prostate biopsies for predicting final Gleason grades in patients who subsequently underwent radical prostatectomy (RP). METHODS: A retrospective review of men who underwent transperineal MR imaging-guided prostate biopsy (tpMRGB) with subsequent radical prostatectomy within 1 year was conducted from 2010 to 2015. All patients underwent a baseline 3-T multiparametric MRI (mpMRI) with endorectal coil and were selected for biopsy based on MR findings of a suspicious prostate lesion and high degree of clinical suspicion for cancer. Spearman correlation was performed to assess concordance between tpMRGB and final RP pathology among patients with and without previous transrectal ultrasound (TRUS)-guided biopsies. RESULTS: A total of 24 men met all eligibility requirements, with a median age of 65 years (interquartile range [IQR] 11.7). The median time from biopsy to RP was 85 days (IQR 50.5). Final pathology revealed Gleason 3 + 4 = 7 in 12 patients, 4 + 3 = 7 in 10 patients, and 4 + 4 = 8 in 2 patients. A strong correlation (ρ: +0.75, p < 0.001) between tpMRGB and RP results was observed, with Gleason scores concordant in 17 cases (71%). 16 of the 24 patients underwent prior TRUS biopsies. Subsequent tpMRGB revealed Gleason upgrading in 88% of cases, which was concordant with RP Gleason scores in 69% of cases (ρ: +0.75, p < 0.001). CONCLUSION: Final Gleason scores diagnosed by tpMRGB at 3-T correlate strongly with final RP surgical pathology. This may facilitate prostate cancer diagnosis, particularly in patients with negative or low-grade TRUS biopsy results in whom clinically significant cancer is suspected or detected on mpMRI.
The basal ganglia is part of a complex system of neuronal circuits that play a key role in the integration and execution of motor, cognitive and emotional function in the human brain. Parkinson's disease is a progressive neurological disorder of the motor circuit characterized by tremor, rigidity, and slowness of movement. Deep brain stimulation (DBS) of the subthalamic nucleus and the globus pallidus pars interna provides an efficient treatment to reduce symptoms and levodopa-induced side effects in Parkinson's disease patients. While the underlying mechanism of action of DBS is still unknown, the potential modulation of white matter tracts connecting the surgical targets has become an active area of research. With the introduction of advanced diffusion MRI acquisition sequences and sophisticated post-processing techniques, the architecture of the human brain white matter can be explored in vivo. The goal of this study is to investigate the white matter connectivity between the subthalamic nucleus and the globus pallidus. Two multi-fiber tractography methods were used to reconstruct pallido-subthalamic, subthalamo-pallidal and pyramidal fibers in five healthy subjects datasets of the Human Connectome Project. The anatomical accuracy of the tracts was assessed by four judges with expertise in neuroanatomy, functional neurosurgery, and diffusion MRI. The variability among subjects was evaluated based on the fractional anisotropy and mean diffusivity of the tracts. Both multi-fiber approaches enabled the detection of complex fiber architecture in the basal ganglia. The qualitative evaluation by experts showed that the identified tracts were in agreement with the expected anatomy. Tract-derived measurements demonstrated relatively low variability among subjects. False-negative tracts demonstrated the current limitations of both methods for clinical decision-making. Multi-fiber tractography methods combined with state-of-the-art diffusion MRI data have the potential to help identify white matter tracts connecting DBS targets in functional neurosurgery intervention.
AIM: To assess single-breath-hold combined positron-emission tomography/computed tomography (PET/CT) for accuracy of tumour image registration and projected ablation volume overlap in patients undergoing percutaneous PET/CT-guided tumour-ablation procedures under general anaesthesia. MATERIALS AND METHODS: Eight patients underwent 12 PET/CT-guided tumour-ablation procedures to treat 20 tumours in the lung, liver, or adrenal gland. Using breath-hold PET/CT, the centre of the tumour was marked on each PET and CT acquisition by four readers to assess two- (2D) and three-dimensional (3D) spatial misregistration. Overlap of PET and CT projected ablation volumes were compared using the Dice similarity coefficient (DSC). Interobserver differences were assessed with repeated measure analysis of variance (ANOVA). Technical success and local progression rates were noted. RESULTS: Mean tumour 2D PET/CT misregistrations were 1.02 mm (range 0.01-5.02), 1.89 (0.03-7.85), and 3.05 (0-10) in the x, y, and z planes. Mean 3D misregistration was 4.4 mm (0.36-10.74). Mean projected PET/CT ablation volume DSC was 0.72 (±0.19). No significant interobserver differences in 3D misregistration (p=0.73) or DSC (p=0.54) were observed. Technical success of ablations was 100%; one (5.3%) of 19 tumours progressed. CONCLUSION: Accurate spatial registration of tumours and substantial overlap of projected ablation volumes are achievable when comparing PET and CT acquisitions from single-breath-hold PET/CT. The results suggest that tumours visible only at PET could be accurately targeted and ablated using this technique.
We propose a method for the automated identification of key white matter fiber tracts for neurosurgical planning, and we apply the method in a retrospective study of 18 consecutive neurosurgical patients with brain tumors. Our method is designed to be relatively robust to challenges in neurosurgical tractography, which include peritumoral edema, displacement, and mass effect caused by mass lesions. The proposed method has two parts. First, we learn a data-driven white matter parcellation or fiber cluster atlas using groupwise registration and spectral clustering of multi-fiber tractography from healthy controls. Key fiber tract clusters are identified in the atlas. Next, patient-specific fiber tracts are automatically identified using tractography-based registration to the atlas and spectral embedding of patient tractography. Results indicate good generalization of the data-driven atlas to patients: 80% of the 800 fiber clusters were identified in all 18 patients, and 94% of the 800 fiber clusters were found in 16 or more of the 18 patients. Automated subject-specific tract identification was evaluated by quantitative comparison to subject-specific motor and language functional MRI, focusing on the arcuate fasciculus (language) and corticospinal tracts (motor), which were identified in all patients. Results indicate good colocalization: 89 of 95, or 94%, of patient-specific language and motor activations were intersected by the corresponding identified tract. All patient-specific activations were within 3mm of the corresponding language or motor tract. Overall, our results indicate the potential of an automated method for identifying fiber tracts of interest for neurosurgical planning, even in patients with mass lesions.
OBJECTIVE: We report nine consecutive percutaneous image-guided cryoablation procedures of head and neck tumors in seven patients (four men and three women; mean age, 68 years; age range, 50-78 years). Ablation of the entire tumor for local control or ablation of a region of tumor for pain relief or preservation of function was achieved in eight of nine procedures. One patient experienced intraprocedural bradycardia, and another developed a neopharyngeal abscess. There were no deaths, permanent neurologic or functional deficits, vascular complications, or adverse cosmetic sequelae due to the procedures. CONCLUSION: Percutaneous image-guided cryoablation offers a potentially less morbid minimally invasive treatment option than salvage head and neck surgery. The complications that we encountered may be avoidable with increased experience. Further work is needed to continue improving the safety and efficacy of cryoablation of head and neck tumors and to continue expanding the use of cryoablation in patients with head and neck tumors that cannot be treated surgically.
OBJECTIVE: To determine if tumor cell density and percentage of Gleason pattern within an outlined volumetric tumor region of interest (TROI) on whole-mount pathology (WMP) correlate with apparent diffusion coefficient (ADC) values on corresponding TROIs outlined on pre-operative MRI. METHODS: Men with biopsy-proven prostate adenocarcinoma undergoing multiparametric MRI (mpMRI) prior to prostatectomy were consented to this prospective study. WMP and mpMRI images were viewed using 3D Slicer and each TROI from WMP was contoured on the high b-value ADC maps (b0, 1400). For each TROI outlined on WMP, TCD (tumor cell density) and the percentage of Gleason pattern 3, 4, and 5 were recorded. The ADCmean, ADC10th percentile, ADC90th percentile, and ADCratio were also calculated in each case from the ADC maps using 3D Slicer. RESULTS: Nineteen patients with 21 tumors were included in this study. ADCmean values for TROIs were 944.8 ± 327.4 vs. 1329.9 ± 201.6 mm(2)/s for adjacent non-neoplastic prostate tissue (p < 0.001). ADCmean, ADC10th percentile, and ADCratio values for higher grade tumors were lower than those of lower grade tumors (mean 809.71 and 1176.34 mm(2)/s, p = 0.014; 10th percentile 613.83 and 1018.14 mm(2)/s, p = 0.009; ratio 0.60 and 0.94, p = 0.005). TCD and ADCmean (ρ = -0.61, p = 0.005) and TCD and ADC10th percentile (ρ = -0.56, p = 0.01) were negatively correlated. No correlation was observed between percentage of Gleason pattern and ADC values. CONCLUSION: DWI MRI can characterize focal prostate cancer using ADCratio, ADC10th percentile, and ADCmean, which correlate with pathological tumor cell density.
PURPOSE: To demonstrate the use of anatomic MRI-visible three-dimensional (3D)-printed phantoms and to assess process accuracy and material MR signal properties. METHODS: A cervical spine model was generated from computed tomography (CT) data and 3D-printed using an MR signal-generating material. Printed phantom accuracy and signal characteristics were assessed using 120 kVp CT and 3 Tesla (T) MR imaging. The MR relaxation rates and diffusion coefficient of the fabricated phantom were measured and (1) H spectra were acquired to provide insight into the nature of the proton signal. Finally, T2 -weighted imaging was performed during cryoablation of the model. RESULTS: The printed model produced a CT signal of 102 ± 8 Hounsfield unit, and an MR signal roughly 1/3(rd) that of saline in short echo time/short repetition time GRE MRI (456 ± 36 versus 1526 ± 121 arbitrary signal units). Compared with the model designed from the in vivo CT scan, the printed model differed by 0.13 ± 0.11 mm in CT, and 0.62 ± 0.28 mm in MR. The printed material had T2 ∼32 ms, T2*∼7 ms, T1 ∼193 ms, and a very small diffusion coefficient less than olive oil. MRI monitoring of the cryoablation demonstrated iceball formation similar to an in vivo procedure. CONCLUSION: Current 3D printing technology can be used to print anatomically accurate phantoms that can be imaged by both CT and MRI. Such models can be used to simulate MRI-guided interventions such as cryosurgeries. Future development of the proposed technique can potentially lead to printed models that depict different tissues and anatomical structures with different MR signal characteristics.
The most recent edition of the prostate imaging reporting and data system (PI-RADS version 2) was developed based on expert consensus of the international working group on prostate cancer. It provides the minimum acceptable technical standards for MR image acquisition and suggests a structured method for multiparametric prostate MRI (mpMRI) reporting. T1-weighted, T2-weighted (T2W), diffusion-weighted (DWI), and dynamic contrast-enhanced (DCE) imaging are the suggested sequences to include in mpMRI. The PI-RADS version 2 scoring system enables the reader to assess and rate all focal lesions detected at mpMRI to determine the likelihood of a clinically significant cancer. According to PI-RADS v2, a lesion with a Gleason score ≥7, volume >0.5 cc, or extraprostatic extension is considered clinically significant. PI-RADS v2 uses the concept of a dominant MR sequence based on zonal location of the lesion rather than summing each component score, as was the case in version 1. The dominant sequence in the peripheral zone is DWI and the corresponding apparent diffusion coefficient (ADC) map, with a secondary role for DCE in equivocal cases (PI-RADS score 3). For lesions in the transition zone, T2W images are the dominant sequence with DWI/ADC images playing a supporting role in the case of an equivocal lesion.
Neurosurgery makes use of preoperative imaging to visualize pathology, inform surgical planning, and evaluate the safety of selected approaches. The utility of preoperative imaging for neuronavigation, however, is diminished by the well-characterized phenomenon of brain shift, in which the brain deforms intraoperatively as a result of craniotomy, swelling, gravity, tumor resection, cerebrospinal fluid (CSF) drainage, and many other factors. As such, there is a need for updated intraoperative information that accurately reflects intraoperative conditions. Since 1982, intraoperative ultrasound has allowed neurosurgeons to craft and update operative plans without ionizing radiation exposure or major workflow interruption. Continued evolution of ultrasound technology since its introduction has resulted in superior imaging quality, smaller probes, and more seamless integration with neuronavigation systems. Furthermore, the introduction of related imaging modalities, such as 3-dimensional ultrasound, contrast-enhanced ultrasound, high-frequency ultrasound, and ultrasound elastography, has dramatically expanded the options available to the neurosurgeon intraoperatively. In the context of these advances, we review the current state, potential, and challenges of intraoperative ultrasound for brain tumor resection. We begin by evaluating these ultrasound technologies and their relative advantages and disadvantages. We then review three specific applications of these ultrasound technologies to brain tumor resection: (1) intraoperative navigation, (2) assessment of extent of resection, and (3) brain shift monitoring and compensation. We conclude by identifying opportunities for future directions in the development of ultrasound technologies.
Significant efforts have been dedicated to minimizing invasiveness associated with surgical interventions, most of which have been possible thanks to the developments in medical imaging, surgical navigation, visualization and display technologies. Image-guided interventions have promised to dramatically change the way therapies are delivered to many organs. However, in spite of the development of many sophisticated technologies over the past two decades, other than some isolated examples of successful implementations, minimally invasive therapy is far from enjoying the wide acceptance once envisioned. This paper provides a large-scale overview of the state-of-the-art developments, identifies several barriers thought to have hampered the wider adoption of image-guided navigation, and suggests areas of research that may potentially advance the field.
OBJECTIVE: Magnetic resonance imaging (MRI) is a widely used imaging modality for studies of knee osteoarthritis (OA). Compared to radiography, MRI offers exceptional soft tissue imaging and true three-dimensional (3D) visualization. However, MRI is expensive both due to the cost of acquisition and evaluation of the images. The goal of our study is to develop a new method to address the cost of MRI by combining innovative acquisition methods and automated post-processing software. METHODS: Ten healthy volunteers were scanned with three different MRI protocols: A standard 3D dual-echo steady state (DESS) pulse sequence, an accelerated DESS (DESS), acquired at approximately half the time compared to DESS, and a multi-echo time DESS (DESS), which is capable of producing measurements of T2 relaxation time. A software tool was used to measure cartilage volume. Accuracy was quantified by comparing DESS to DESS and DESS and precision was measured using repeat readings and acquisitions. T2 precision was determined using duplicate DESS acquisitions. Intra-class correlation coefficients (ICCs), root-mean square standard deviation (RMSSD), and the coefficient of variation (CoV) were used to quantify accuracy and precision. RESULTS: The accuracies of DESS and DESS were CoV = 3.7% and CoV = 6.6% respectively, while precision was 3.8%, 3.0%, and 3.1% for DESS, DESS and DESS. T2 repositioning precision was 5.8%. CONCLUSION: The results demonstrate that accurate and precise quantification of cartilage volume is possible using a combination of substantially faster MRI acquisition and post-processing software. Precise measurements of cartilage T2 and volume can be made using the same acquisition.
BACKGROUND: Bilateral mastectomies (BM) are traditionally performed by single surgeons (SS); a co-surgeon (CS) technique, where each surgeon concurrently performs a unilateral mastectomy, offers an alternative approach. We examined differences in general surgery time (GST), overall surgery time (OST), and patient complications for BM performed by CS and SS. METHODS: Patients undergoing BM with tissue expander reconstruction (BMTR) between January 2010 and May 2014 at our center were identified through operative case logs. GST (incision to end of BM procedure), reconstruction duration (RST) (plastic surgery start to end of reconstruction) and OST (OST = GST + RST) was calculated. Patient age, presence/stage of cancer, breast weight, axillary procedure performed, and 30-day postoperative complications were extracted from medical records. Differences in GST and OST between CS and SS cases were assessed with a t test. A multivariate linear regression was fit to identify factors associated with GST. RESULTS: A total of 116 BMTR cases were performed [CS, n = 67 (57.8 %); SS, n = 49 (42.2 %)]. Demographic characteristics did not differ between groups. GST and OST were significantly shorter for CS cases, 75.8 versus 116.8 min, p < .0001, and 255.2 versus 278.3 min, p = .005, respectively. Presence of a CS significantly reduces BMTR time (β = -38.82, p < .0001). Breast weight (β = 0.0093, p = .03) and axillary dissection (β = 28.69, p = .0003) also impacted GST. CONCLUSIONS: The CS approach to BMTR reduced both GST and OST; however, the degree of time savings (35.1 and 8.3 %, respectively) was less than hypothesized. A larger study is warranted to better characterize time, cost, and outcomes of the CS-approach for BM.
PURPOSE: To compare clinical outcomes of image-based versus non-image-based interstitial brachytherapy (IBBT) for vaginal cancer. METHODS AND MATERIALS: Of 72 patients with vaginal cancer treated with brachytherapy (BT), 47 had image guidance (CT=31, MRI=16) and 25 did not. Kaplan-Meier (KM) estimates were generated for any recurrence, local control (LC), disease-free interval (DFI), and overall survival (OS) and Cox models were used to assess prognostic factors. RESULTS: Median age was 66 and median follow-up time was 24months. Median cumulative EQD2 dose was 80.8Gy in the non-IBBT group and 77Gy in the IBBT group. For non-IBBT versus IBBT, the 2-year KM LC was 71% vs. 93% (p=0.03); DFI was 54% vs. 86% (p=0.04); and OS 52% vs. 82% (p=0.35). On multivariate analysis, IBBT was associated with better DFI (HR 0.24, 95% CI 0.07-0.73). Having any 2 or more of chemotherapy, high-dose-rate (HDR) BT or IBBT (temporally correlated variables) significantly reduced risk of relapse (HR=0.33, 95% CI=0.13-0.83), compared to having none of these factors. CONCLUSION: Over time, the use of chemotherapy, HDR, and IBBT has increased in vaginal cancer. The combination of these factors resulted in the highest rates of disease control. Image-guided brachytherapy for vaginal cancer patients maximizes disease control.
We report percutaneous cryoablation of spine tumors in 7 consecutive patients (5 men, 2 women [mean age, 47 years; range, 17-68 years]) by using intraprocedural image monitoring of ice ball margins to protect adjacent neural elements. Complete tumor ablation was achieved in all patients without neurologic complication. Pain relief was achieved in 4 of 5 (80%) patients; the patient with persistent pain was later found to have enlarging metastases at other sites.
OBJECTIVE: Accurate biopsy sampling of the suspected lesions is critical for the diagnosis and clinical management of prostate cancer. Transperineal in-bore MRI-guided prostate biopsy (tpMRgBx) is a targeted biopsy technique that was shown to be safe, efficient, and accurate. Our goal was to develop an open source software platform to support evaluation, refinement, and translation of this biopsy approach. METHODS: We developed SliceTracker, a 3D Slicer extension to support tpMRgBx. We followed modular design of the implementation to enable customization of the interface and interchange of image segmentation and registration components to assess their effect on the processing time, precision, and accuracy of the biopsy needle placement. The platform and supporting documentation were developed to enable the use of software by an operator with minimal technical training to facilitate translation. Retrospective evaluation studied registration accuracy, effect of the prostate segmentation approach, and re-identification time of biopsy targets. Prospective evaluation focused on the total procedure time and biopsy targeting error (BTE). RESULTS: Evaluation utilized data from 73 retrospective and ten prospective tpMRgBx cases. Mean landmark registration error for retrospective evaluation was 1.88 ± 2.63 mm, and was not sensitive to the approach used for prostate gland segmentation. Prospectively, we observed target re-identification time of 4.60 ± 2.40 min and BTE of 2.40 ± 0.98 mm. CONCLUSION: SliceTracker is modular and extensible open source platform for supporting image processing aspects of the tpMRgBx procedure. It has been successfully utilized to support clinical research procedures at our site.
Patient-mounted needle guide devices for percutaneous ablation are vulnerable to patient motion. The objective of this study is to develop and evaluate a software system for an MRI-compatible patient-mounted needle guide device that can adaptively compensate for displacement of the device due to patient motion using a novel image-based automatic device-to-image registration technique. We have developed a software system for an MRI-compatible patient-mounted needle guide device for percutaneous ablation. It features fully-automated image-based device-to-image registration to track the device position, and a device controller to adjust the needle trajectory to compensate for the displacement of the device. We performed: (a) a phantom study using a clinical MR scanner to evaluate registration performance; (b) simulations using intraoperative time-series MR data acquired in 20 clinical cases of MRI-guided renal cryoablations to assess its impact on motion compensation; and (c) a pilot clinical study in three patients to test its feasibility during the clinical procedure. FRE, TRE, and success rate of device-to-image registration were [Formula: see text] mm, [Formula: see text] mm, and 98.3% for the phantom images. The simulation study showed that the motion compensation reduced the targeting error for needle placement from 8.2 mm to 5.4 mm (p < 0.0005) in patients under general anesthesia (GA), and from 14.4 mm to 10.0 mm ([Formula: see text]) in patients under monitored anesthesia care (MAC). The pilot study showed that the software registered the device successfully in a clinical setting. Our simulation study demonstrated that the software system could significantly improve targeting accuracy in patients treated under both MAC and GA. Intraprocedural image-based device-to-image registration was feasible.
PURPOSE: To develop and evaluate an approach to estimate the respiratory-induced motion of lesions in the chest and abdomen. MATERIALS AND METHODS: The proposed approach uses the motion of an initial reference needle inserted into a moving organ to estimate the lesion (target) displacement that is caused by respiration. The needles position is measured using an inertial measurement unit (IMU) sensor externally attached to the hub of an initially placed reference needle. Data obtained from the IMU sensor and the target motion are used to train a learning-based approach to estimate the position of the moving target. An experimental platform was designed to mimic respiratory motion of the liver. Liver motion profiles of human subjects provided inputs to the experimental platform. Variables including the insertion angle, target depth, target motion velocity and target proximity to the reference needle were evaluated by measuring the error of the estimated target position and processing time. RESULTS: The mean error of estimation of the target position ranged between 0.86 and 1.29 mm. The processing maximum training and testing time was 5 ms which is suitable for real-time target motion estimation using the needle position sensor. CONCLUSION: The external motion of an initially placed reference needle inserted into a moving organ can be used as a surrogate, measurable and accessible signal to estimate in real-time the position of a moving target caused by respiration; this technique could then be used to guide the placement of subsequently inserted needles directly into the target.
Brain shift during tumor resection compromises the spatial validity of registered preoperative imaging data that is critical to image-guided procedures. One current clinical solution to mitigate the effects is to reimage using intraoperative magnetic resonance (iMR) imaging. Although iMR has demonstrated benefits in accounting for preoperative-to-intraoperative tissue changes, its cost and encumbrance have limited its widespread adoption. While iMR will likely continue to be employed for challenging cases, a cost-effective model-based brain shift compensation strategy is desirable as a complementary technology for standard resections. We performed a retrospective study of [Formula: see text] tumor resection cases, comparing iMR measurements with intraoperative brain shift compensation predicted by our model-based strategy, driven by sparse intraoperative cortical surface data. For quantitative assessment, homologous subsurface targets near the tumors were selected on preoperative MR and iMR images. Once rigidly registered, intraoperative shift measurements were determined and subsequently compared to model-predicted counterparts as estimated by the brain shift correction framework. When considering moderate and high shift ([Formula: see text], [Formula: see text] measurements per case), the alignment error due to brain shift reduced from [Formula: see text] to [Formula: see text], representing [Formula: see text] correction. These first steps toward validation are promising for model-based strategies.
OBJECTIVE: The purpose of this article is to report our intermediate to long-term outcomes with image-guided percutaneous hepatic tumor cryoablation and to evaluate its technical success, technique efficacy, local tumor progression, and adverse event rate. MATERIALS AND METHODS: Between 1998 and 2014, 299 hepatic tumors (243 metastases and 56 primary tumors; mean diameter, 2.5 cm; median diameter, 2.2 cm; range, 0.3-7.8 cm) in 186 patients (95 women; mean age, 60.9 years; range, 29-88 years) underwent cryoablation during 236 procedures using CT (n = 126), MRI (n = 100), or PET/CT (n = 10) guidance. Technical success, technique efficacy at 3 months, local tumor progression (mean follow-up, 2.5 years; range, 2 months to 14.6 years), and adverse event rates were calculated. RESULTS: The technical success rate was 94.6% (279/295). The technique efficacy rate was 89.5% (231/258) and was greater for tumors smaller than 4 cm (93.4%; 213/228) than for larger tumors (60.0%; 18/30) (p < 0.0001). Local tumor progression occurred in 23.3% (60/258) of tumors and was significantly more common after the treatment of tumors 4 cm or larger (63.3%; 19/30) compared with smaller tumors (18.0%; 41/228) (p < 0.0001). Adverse events followed 33.8% (80/236) of procedures and were grade 3-5 in 10.6% (25/236) of cases. Grade 3 or greater adverse events more commonly followed the treatment of larger tumors (19.5%; 8/41) compared with smaller tumors (8.7%; 17/195) (p = 0.04). CONCLUSION: Image-guided percutaneous cryoablation of hepatic tumors is efficacious; however, tumors smaller than 4 cm are more likely to be treated successfully and without an adverse event.